To avoid damage of the connector housing when rectangular or so-called
spade-headed tip-styles are used, it is eminent to ensure that the homeposition
of the Probe does not change. Therefore, non-rotating Probes
are designed in such a way that the plunger head is guided by force and
subsequently rotating is prevented.
Solely when the Probe is pressed/screwed into the Receptacle is it necessary
to set the desired position of the plunger head.
|Series||Grid||Working Stroke in mm||Max. Stroke||Current Rating (A)|
|GKS-710||100||4||5||5 - 8|
|GKS-803 M||177||6,4||8||5 - 15|
|GKS-098||200||4||5||8 - 10|
|GKS-714||200||4 / 6||5 / 7||8 - 10|
|GKS-746 M||100||4||5||5 - 8|